Size exclusion chromatography
|Size exclusion chromatography|
Equipment for running size exclusion chromatography. The buffer is pumped through the column (right) by a computer controlled device
|Related||High performance liquid chromatography|
Aqueous Normal Phase Chromatography
Ion exchange chromatography
Micellar liquid chromatography
Size exclusion chromatography (SEC) is a chromatographic method in which particles are separated based on their size, or in more technical terms, their hydrodynamic volume. It is usually applied to large molecules or macromolecular complexes such as proteins and industrial polymers. When an aqueous solution is used to transport the sample through the column, the technique is known as gel filtration chromatography. The name gel permeation chromatography is used when an organic solvent is used as a mobile phase. The main application of gel filtration chromatography is the fractionation of proteins and other water-soluble polymers, while gel permeation chromatography is used to analyze the molecular weight distribution of organic-soluble polymers. Either technique should not be confused with gel electrophoresis, where an electric field is used to "pull" or "push" molecules through the gel depending on their electrical charges.
SEC is a widely used technique for the purification and analysis of synthetic and biological polymers, such as proteins, polysaccharides and nucleic acids. Biologists and biochemists typically use a gel medium — usually polyacrylamide, dextran or agarose — and filter under low pressure. Polymer chemists typically use either a silica or crosslinked polystyrene medium under a higher pressure. These media are known as the stationary phase.
The advantage of this method is that the various solutions can be applied without interfering with the filtration process, while preserving the biological activity of the particles to be separated. The technique is generally combined with others that further separate molecules by other characteristics, such as acidity, basicity, charge, and affinity for certain compounds.
The technique was invented by Grant Henry Lathe and Colin R Ruthven, working at Queen Charlotte’s Hospital, London . They later received the John Scott Award for this invention . While Lathe and Ruthven used starch gels as the matrix, Porath and Flodin later introduced dextran gels ; other gels with size fractionation properties include agarose and polyacrylamide. A short review of these developments has appeared .
Theory and method
The underlying principle of SEC is that particles of different sizes will elute (filter) through a stationary phase at different rates. This results in the separation of a solution of particles based on size. Provided that all the particles are loaded simultaneously or near simultaneously, particles of the same size should elute together.
This is usually achieved with an apparatus called a column, which consists of a hollow tube tightly packed with extremely small porous polymer beads designed to have pores of different sizes. These pores may be depressions on the surface or channels through the bead. As the solution travels down the column some particles enter into the pores. Larger particles cannot enter into as many pores. The larger the particles, the less overall volume to traverse over the length of the column, and the faster the elution.
The filtered solution that is collected at the end is known as the eluate. The void volume includes any particles too large to enter the medium, and the solvent volume is known as the column volume.
Factors affecting filtration
In real life situations particles in solution do not have a constant, fixed size, resulting in the probability that a particle which would otherwise be hampered by a pore may pass right by it. Also, the stationary phase particles are not ideally defined; both particles and pores may vary in size. Elution curves therefore resemble Gaussian distributions. The stationary phase may also interact in undesirable ways with a particle and influence retention times, though great care is taken by column manufacturers to use stationary phases which are inert and minimize this issue.
Like other forms of chromatography, increasing the column length will enhance the resolution, and increasing the column diameter increases the capacity of the column. Proper column packing is important to maximize resolution: an overpacked column can collapse the pores in the beads, resulting in a loss of resolution. An underpacked column can reduce the relative surface area of the stationary phase accessible to smaller species, resulting in those species spending less time trapped in pores.
In simple manual columns the eluent is collected in constant volumes, known as fractions. The more similar the particles are in size, the more likely they will be in the same fraction and not detected separately. More advanced columns overcome this problem by constantly monitoring the eluent.
The collected fractions are often examined by spectroscopic techniques to determine the concentration of the particles eluted. Three common spectroscopy detection techniques are refractive index (RI), evaporative light scattering (ELS), and ultraviolet (UV). When eluting spectroscopically similar species (such as during biological purification) other techniques may be necessary to identify the contents of each fraction. The elution volume decreases roughly linearly with the logarithm of the molecular hydrodynamic volume (often assumed to be proportional to molecular weight). Columns are often calibrated using 4-5 standard samples (e.g., folded proteins of known molecular weight) to determine the void volume and the slope of the logarithmic dependence. This calibration may need to be repeated under different solution conditions.
It is also possible to analyse the eluent flow continuously with RI, LALLS UV and or viscosity measurements.
SEC is generally considered a low resolution chromatography as it does not discern similar species very well, and is therefore often reserved for the final "polishing" step of a purification. The technique can determine the quaternary structure of purified proteins which have slow exchange times, since it can be carried out under native solution conditions, preserving macromolecular interactions. SEC can also assay protein tertiary structure as it measures the hydrodynamic volume (not molecular weight), allowing folded and unfolded versions of the same protein to be distinguished. For example, the apparent hydrodynamic radius of a typical protein domain might be 14 Å and 36 Å for the folded and unfolded forms respectively. SEC allows the separation of these two forms as the folded form will elute much later due to its smaller size. Alternatively, folded and unfolded versions of the same metalloproteins can be separated according to their different isoelectric points by using quantitative preparative native continuous polyacrylamide gel electrophoresis (QPNC-PAGE).
SEC can be used as a measure of both the size and the polydispersity of a synthesised polymer - that is, the ability to be able to find the distribution of the sizes of polymer molecules. If standards of a known size are run previously, then a calibration curve can be created to determine the sizes of polymer molecules of interest. Alternatively, techniques such as light scattering and/or viscometry can be used online with SEC to yield absolute molecular weights that do not rely on calibration with standards of known molecular weight. Due to the difference in size of two polymers with identical molecular weights, the absolute determination methods are generally more desirable. A typical SEC system can quickly (in about half an hour) give polymer chemists information on the size and polydispersity of the sample.
- Lathe, GH and Ruthven, CR (1955) The separation of substances on the basis of their molecular weights, using columns of starch and water. Biochem J. 60(4):xxxiv.
- Lathe, GH and Ruthven, CR (1956) The separation of substances and estimation of their relative molecular sizes by the use of columns of starch in water. Biochem. J. 62(4): 665-674. article
- John Scott Award
- Porath, J and Flodin, P (1959) Gel filtration: A method for desalting and group separation. Nature 183(4676): 1657-1659.
- Eisenstein, M (2006) A look back, adventures in the matrix. Nature Methods 3(5): 410 article
ar:تفريق لوني بعبور الهلام bs:Ekskluziona hromatografija cs:Gelová chromatografie de:Gel-Permeations-Chromatografie it:Cromatografia a permeazione di gel nl:Gelpermeatiechromatografie sk:Gélová permeačná chromatografia sv:Gelfiltrering uk:Ексклюзійна хроматографія